The STYLCAM 200 R, a Rotary Tablet Press Simulator - Optimising the Instrumentation: Characterising the Displacement Measurement System
Oct. 25, 2008, midnight
R F Lammens
Remove from favorites
Add to favorites
It is possible to measure the densification part of “force-displacement” curves with sufficient accuracy,because the relation between core position and output signal of the displacement measurement system being used is far more better than their
specified precision. Due to the excellent reproducible deviation from linearity, a polynomial regression may be used for discribing this relation. For the examined displacement transducers, it is necessary to prevent the plunger from rotation.